I have an Interro PDA 2000 lab scope. I love the tool, but it gave up the ghost last Friday. It will turn on and go through a self check, then go to the main screen. As soon as I touch any button, it beeps and the screen goes blank. If you wait a while, or flip the reset switch, it will start over again, but I can never get it to go past the main screen. Has anyone else encountered this problem? I'm wondering if it is just a bad software card. It looks as though there is nowhere to get these repaired anymore, and I can not find a software card for sale anywhere. The software card says 2100A V.3.1Gas, PC link. Does anyone have any suggestions?
14 Replies Received (View Replies)
14 Replies Received (Hide Replies)